Feb. 28, 2017
Total shoulder replacement has clinical efficacy for the treatment of primary and secondary degenerative conditions of the shoulder. It has excellent surgical outcomes, providing long-term pain relief and improved function for the majority of patients. It is also a demanding procedure, with known complications. Failure of the glenoid component of the implant is one of the most common and worrisome complications in the mid to long term.
Glenoid component failure can result from anatomic factors such as glenoid version or rotator cuff insufficiency, from fixation problems, or from the component itself — for example, metal-backed versus cemented polyethylene glenoids. Inaccurate component positioning is also a critical factor in long-term implant survival; improperly placed implants increase the risk of instability, wear and loosening and the need for revision surgery, which is typically more challenging and less successful than the primary operation.
Proper glenoid placement in terms of version and inclination is difficult, however, especially in the presence of complex and irregular patterns of glenoid bone loss with unreliable landmarks. When reverse arthroplasty is performed in these cases, directing the glenoid baseplate along an axis with enough bone stock for fixation can be a daunting task.
Joaquin Sanchez-Sotelo, M.D., Ph.D., an orthopedic surgeon specializing in shoulder and elbow surgery at Mayo Clinic's campus in Rochester, Minnesota, observes that the classic method of using standard X-rays to estimate areas of missing bone and guide placement of the component in space is imperfect.
"We are often able to get it right, but sometimes the amount of bone loss is so abnormal that it is very difficult," he says. "Fortunately, three developments in the last few years — 3-D preoperative planning, patient-specific instrumentation and 3-D printing — have greatly increased the accuracy of glenoid component placement and improved the likelihood of optimal outcomes in complicated cases."
Three-D computer planning for implant placement originated with early studies investigating whether 3-D imaging helped surgeons better identify glenoid bone loss. In the current process, two-dimensional CT images of a patient's scapula and humerus are uploaded into a software system, which reformats them into accurate 3-D models. This allows far better visualization of the glenoid cavity than conventional X-rays or two-dimensional CT. It also allows surgeons to clearly visualize and consider preservation of support bone, implant version and inclination angle, and glenoid contact for each patient.
A large body of research has demonstrated improved glenoid orientation with the use of preplanning software; most studies demonstrate average derivations in version and inclination of 5 degrees or less from the intended component position.
John W. Sperling, M.D., an orthopedic surgeon specializing in shoulder surgery at Mayo Clinic's campus in Rochester, Minnesota, notes the large volume of complex cases seen at Mayo Clinic has helped drive innovation in this space.
"This is an extremely powerful tool because it allows movement of the component to show varying degrees of orientation. It is especially helpful in cases with marked distortion of anatomy and significant bone loss," he says. "In the four years we have been helping develop this technology at Mayo Clinic, the degree of sophistication has increased substantially, so we can now plan the surgery completely on the computer before entering the operating room."
Surgical planning for PSI
PSI allows surgical implementation of the preoperative plan through the use of patient-specific guides.
PSI for placing the shoulder arthroplasty
PSI for placing the shoulder arthroplasty
Instrumentation directs guidewire
Instrumentation helps direct the guidewire in the correct version and inclination.
Short stem uncemented reverse arthroplasty placement
Patient undergoes placement of a short stem uncemented reverse arthroplasty.
Three-D printed scapula
Patient-specific instrumentation (PSI) allows surgical implementation of the preoperative plan through the use of patient-specific guides that direct glenoid implant placement and fixation, including positioning and orientation of the implant, the reaming depth and angle, and the trajectory of drilling for screw placement. Several vendors offer PSI, and the types of instruments vary considerably, including pin guides versus cutting guides, a greater or lesser degree of surgeon input in the preplanning process, and various types of imaging for the 3-D model. It is unclear whether any is superior, and the choice depends mainly on cost and physician preference.
Many groups have reported on the use of PSI for shoulder arthroplasty, with generally positive results. A randomized controlled trial published in the December 2012 edition of The Journal of Bone & Joint Surgery, American Volume, compared PSI with standard instrumentation for glenoid placement. Overall, the PSI group had fewer incorrectly positioned glenoid components and significant improvement in the selection and use of the appropriate implant for patient anatomy.
3-D printers use computer software to create physical objects from CT data. In the last decade, technological advances have lowered the cost of 3-D printers, thereby expanding their use. Custom 3-D-printed implants have been used in femoral and tibial hemiarthroplasty, and in 2015, surgeons at Mayo Clinic's campus in Arizona performed bilateral total hip arthroplasty using 3-D-printed implants for a patient with a bone growth disorder who was too small for conventional hip implants.
"Mayo Clinic owns 3-D printers, so we can print the bones of the most complicated cases in-house in a day," Dr. Sanchez-Sotelo says. "It's very powerful to have a printed replica of the bone in your hands outside the body while performing the procedure."
Computer planning programs and PSI appear to increase accuracy and efficiency, minimize outliers, and slightly lower operative time and costs. But they have not been proved to be cost-effective and whether the added cost is clinically beneficial in the long-term is a matter of debate.
Dr. Sperling says computer planning and PSI are generally reserved for patients with complex anatomy, including dysplasia, bone deficiency or deformity secondary to prior trauma. "We perform more shoulder replacement surgeries than any other center in the United States and are a leader in PSI for shoulder arthroplasty. In part that is because we have been a pioneer in helping develop PSI techniques and in part because we see many complex cases. For those patients, we have all the tools needed to ensure a successful surgical outcome and implant durability," he says.
For more information
Three-D printed shoulder
Hendel MD, et al. Comparison of patient-specific instruments with standard surgical instruments in determining glenoid component position: A randomized prospective clinical trial. The Journal of Bone & Joint Surgery, American Volume. 2012;94:2167.